HANDLING THIS SERVICE MANUAL About This Service Manual n. Scope This Service Manual applies to the Fuji Medical Dry Laser Imager DRYPIX Service Technicians PACS / DR / CR / Digital Forum Drypix Service Manual CARLOS RUEDA Innovacion Tecnologica Hello, I am trying. View and Download FujiFilm DRYPIX service manual online. DRYPIX Printer pdf manual download.
|Published (Last):||21 February 2008|
|PDF File Size:||9.49 Mb|
|ePub File Size:||11.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
Add to Servics Bench. Evaluation of a computed radiography detector for FFDM separates out the components of image acquisition, Remove the belt cover. Prepare the PC for servicing.
Operation Panel Remove the operation panel. Remove the conveyance driving roller lower.
Page 24 Reinstalling Procedure Reinstall in the reverse order of removal. Remove the special screws. Heat Development Unit Remove the film release unit.
service manual – Fujifilm DryPix | MedWrench
Shutdown the PC for servicing. Disconnect the network cable of the PC for servicing from the network to manuxl the equipment is connected. Power supply is not switching on. Some variation from the published values may be caused by slight differences between the measurements of the internal printer densitometer and the external densitometer used at the site. When installation completed, the [Software Install Finish!! Page 42 Insert pins such as hexagonal wrench into the two holes on gear seervice, and adjust the phase of the removal drive cam.
However, perform the following when securing the sub-scanning unit into the equipment.
FujiFilm DRYPIX 7000 Service Manual
The source of the problem must be corrected before any mammographic images are printed. It accommodates all common film sizes, and has an optional second film drawer and top-mounted 4-bin film sorter.
Tray Loading Unit Steps that differ between the three units are indicated separately. Film Positioning Motor upper md5 Remove the upper conveyor unit.
You can adjust your Community Subscriptions in Settings. Remove the PC board. Remove the heat development unit. Geometry Recording Length and Width 4. Page 16 MC Remove the tray inside cover.
Filter Section Cooling Fan Remove the cooling fan assembly.
Page 88 35 x 43cm size 26 x 36 cm size 20 x 25 cm size To print the manual completely, please, download it. Page MC 7. Page 48 To return the film size from 20 x 25 cm to other sizes, refer to the following figure and change the attaching position of the film release plate.
DryPix Plus (4000) QC for Mammography
MC Remove the suction cup arm. Control limit Step 1 Step 5 Step 10 Step Remove Subscription Confirm Remove. Serivce the grid pattern using the M-Utility. Remove the lower conveyor unit. Thank you much in advance Reply. Cleaning Roller Grip Release Motor md4 Rotate the worm gear of the cleaning roller grip release motor MD4 in the arrow direction shown in the figure with the hand so that it grips the cleaning roller.
Due to advances in technology, the regulation and testing of Page Reinstall in the reverse order of removal. EPS SP Mechanical Jamming, Abnormal Sound 2. Page 52 Push down the plate, and with an approximately 50 mm space between the plate and stopper, push the suction cup manuall the film.
Page 28 Place the tip in front and behind the right rail of the tray loading unit on the aperture of the tray loading unit, tighten the rail temporarily with the screws. Page 71 Open the left and right positioning guides until they touch the sides when installing the film positioning motor MD5. Page When reinstalling the adiabatic cover, do not sandwich the thermistor cable. By pressing ‘print’ button you will print only current page.
EPS Change the attaching position of the film release plate. Film is produced in batches. Test Contents Mechanical Output test pattern films to make sure that there are no mechanical transportation errors or abnormal sounds.
Fujifilm – DryPix Community, Manuals and Specifications | MedWrench
Geometry Output test pattern films and measure the specified horizontal and vertical length to confirm it is within specifications. If problems are detected, corrective action dfypix be taken before clinical films are processed under less than optimal conditions.
After using the manual, keep it nearby for future reference. As there is problem in power supply of drypix would like to have its schematic diagram.